Année de Publication : 2008
Type : Article
Thème : Chimie de l’environnement
Experimental and theoretical uncertainties in the measurement of cloud condensation nuclei (CCN) with a continuous-flow thermal-gradient CCN counter from Droplet Measurement Technologies (DMT-CCNC) have been assessed by model calculations and calibration experiments with ammonium sulfate and sodium chloride aerosol particles in the diameter range of 20–220 nm. Experiments have been performed in the laboratory and during field measurement campaigns, covering a wide range of instrument operating conditions (650–1020 hPa pressure, 293–303K inlet temperature, 4–34Km−1 temperature gradient, 0.5–1.0 L min−1 flow rate). For each set of conditions, the effective water vapor supersaturation (Seff, 0.05–1.4%) was determined from the measured CCN activation spectra (dry particle activation diameters) and K¨ohler model calculations. High measurement precision was achieved under stable laboratory conditions, where the relative standard deviations of Seff were as low as ±1%. During field measurements, however, the relative deviations increased to about ±5%, which can be mostly attributed to variations of the CCNC column top temperature with ambient temperature. The observed dependence of Seff on temperature, pressure, and flow rate was compared to the CCNC flow model of Lance et al. (2006). At high Seff the relative deviations between flow model and experimental results were mostly less than 10%, but at Seff_0.1% they exceeded 40%. Thus, careful experimental calibration is required for high-accuracy CCN measurements – especially at low Seff. A comprehensive comparison and uncertainty analysis of the various K¨ohler models and thermodynamic parameterizations commonly used in CCN studies showed that the relative deviations between different approaches are as high as 25% for (NH4)2SO4 and 12% for NaCl. The deviations were mostly caused by the different parameterizations for the activity of water in aqueous solutions of the two salts. To ensure comparability of results, we suggest that CCN studies should always report exactly which K¨ohler model equations and parameters were used. Provided that the Aerosol Inorganics Model (AIM) can be regarded as an accurate source of water activity data for highly dilute solutions of (NH4)2SO4 and NaCl, only K¨ohler models that are based on the AIM or yield similar results should be used in CCN studies involving these salts and aiming at high accuracy. Experiments with (NH4)2SO4 and NaCl aerosols showed that the conditions of particle generation and the shape and microstructure of NaCl particles are critical for their application in CCN activation experiments (relative deviations up to 18%).